ZE102 ESD Wrist Strap Tester
ESD Wrist Strap Tester is designed for fast, frequent testing of ESD wrist straps.
It is very simple to operate.
A green light signals the user that everything is OK. A red light and an audible indicator mean that the circuit resistance is either too low or too high.
ESD Wrist Strap is the first lines of defense against electrostatic charge build up on the human body. They are designed to carry away this charge as soon as it is generated.
At the same time the wrist strap or footwear must remain electrically safe for the user. Thus, it is important to test every wrist strap regularly.
In order to protect sensitive electronic components, the maximum wrist strap resistance should not be more than 10 MΩ.
On the other hand, the path to ground resistance should not be less than 0.75 MΩ as protection to the wearer from dangerous levels of voltage and current flow.
The ESD Wrist Strap Tester checks these parameters under actual operating conditions with the strap on the wrist.
It clearly indicates whether or not the resistance of the strap system including the wearer is between the safe limits 750 KΩ to 10 MΩ. If you are looking for ESD wrist strap & footwear tester, then ZE103 can be an option available with us.
|Pass Band||Wrist Strap 0.75 to 10.0 MΩ|
||By Pressing Metal Button
Dual Color LED :
Pass - Green, Fail - Red
Audible Alarm for Fail mode
|Dimensions||(L)135 mm x (W)70 mm x (H)25 mm|
|| To National Standards
||9 V DC Battery
|Low Battery||Orange Warning at less than 7 V|
||Recommended every 12 months
||EN 100 015|
|Country of Origin||India|
1. Power the instrument for use by opening the back cover to install a 9 V battery.
2. For Checking Wrist Strap:
2.1 Connect the wrist strap cord to the instrument into the socket provided and with the strap on the wrist.
2.2 Press gently on the metal sensor on the instrument.
2.3 The instrument will check whether the resistance of the wrist strap falls within the acceptable range (0.75 to 10.0 MΩ).
2.4 This is indicated by the LED lighting up Green.
|Apart from routine battery replacement, as and when indicated by low battery indication, this instrument is designed to be completely maintenance free.
There are no user serviceable parts inside. In case of malfunction, calibration refer to the authorized service centre or personnel.
For cleaning the instrument, use a damp cloth, never use any solvent-based cleaners as they may damage the plastic case.
* Warranty is applicable from the date of invoice. Repair warranty only for the instrument. Forward-Reverse transportation cost from our office in Delhi, India will be paid by buyer of the equipment. Warranty voids in case of liquid damage, physical damage and/or if warranty seal is voided off.
|Battery 9 V
|Download Technical Datasheet|
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.